AIRANGER XPL PLUS PDF
This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||3 July 2006|
|PDF File Size:||1.37 Mb|
|ePub File Size:||13.44 Mb|
|Price:||Free* [*Free Regsitration Required]|
Make offer – Loading Mouse over to zoom – Click to enlarge. Add to basket. Scroll or direct access the desired Breakpoint. Comments to this Manuals Your Name. ALL” is displayed until the reset is complete. This feature is preset to 0.
Wipe out the area of the enclosure lid docking bay recess with a clean dry cloth if necessary before installing the programmer. Profile Records p To P Number field. Generally, the more breakpoints entered, the greater the volume calculation accuracy.
Before using this feature, verify the: To unLock, direct access this parameter and enter the value “”. Learn More – opens in a new window or tab Any international postage is paid in part to Pitney Bowes Inc.
Otherwise, when “fastback” is selected, the Failsafe Level Advance is restricted, however the advance to the new material level when a valid measurement is made is immediate.
Milltronics airanger XPL plus Instruction Manual
If the temperature varies with distance from the transducer, enter the average temperature. However, it can be reduced under certain circum- stances. Proceed to Measurement Difficul- ties. To reference a material level to sea level, enter the distance in Units Pbetween Empty P and sea level.
MILLTRONICS AIRANGER XPL PLUS INSTRUCTION MANUAL Pdf Download.
Not all components or their maximum quantity may be required. If the Reading is currently displayed in feet, to display in yards, enter 3. May not post to United States – Read item description or contact seller for postage options. Application Assistance The preceding examples describe only a few ways in which the XPL Plus can be applied to process measurement requirements. Page 80 Enter the point number to copy from. There are 1 items available.
Milltronics AiRanger XPL Plus Instruction Manual
An XPL Plus level monitor, utilizing a single Siemens Milltronics ultrasonic transducer, accurately monitors vessel material level without material contact. Select “short and long” to have short and long acoustic shots fired each time a vessel is scanned. Relays may be programmed as Standard Alarms or for Custom Relay operation. Loosen the 6 enclosure lid captivated screws and swing the lid open.
Transducer sound beam is free from obstruction. Based on level, space, distance, or volume, Immune to power interruptions, all programming is stored indefinitely. Range Calibration Parameters p To P 2.
The Profile Pointer may be moved to a number of points on the Echo Profile, to gain specific information dependent upon the Profile Pointer Pxl used. Page 68 Extend Near Blanking to overcome measurement difficulties which cannot be corrected by transducer location, mounting, or aiming.
Don’t have an account? The XPL Plus will perform reliably, with little or no maintenance.
Siemens Milltronics AiRanger XPL Plus Multipoint Level Transmitter System
Parameter Value in memory complete a parameter reset. Wrong Reading If the material monitored is a liquid, check for splashing in the vessel. Removed from a fuel distribution depot for upgrading purposes. If material build-up on the transducer face is observed, to maintain maximum system performance, a schedule of regular cleaning should be adopted.